INDICATIONS FOR USE
The Hydrophilic Guidewire is indicated for general intravascular use to aid in the selective placement of catheters in the peripheral and cerebral vasculature during diagnostic and/or therapeutic procedures. The device is not intended for use in the coronary arteries.
The Guidewire Torque Device is intended to facilitate guidewire manipulation during interventional procedures.
The Guidewire Introducer is intended to facilitate guidewire insertion into a catheter hub or through a hemostasis valve.
- Federal (USA) law restricts this device to sale by or on the order of a physician.
- This device should be used only by physicians with a thorough understanding of angiography and percutaneous interventional radiologic procedures.
- Do not use if pouch is open or damaged.
- Prior to use, carefully examine the guidewire and its packaging to verify no damage has occurred during shipment. Do not use if pouch is open or damaged.
- Inspect the guidewire for any surface irregularities, bends or kinks. Any guidewire damage may decrease the desired performance characteristics. Avoid exposure of the guidewire coating to disinfection agents such as alcohols. This may result in coating delamination.
- The hydrophilic coating bonded to the guidewire is an extremely lubricious surface when properly hydrated. Ensure proper hydration by allowing entire guidewire to become wet for at least 30 seconds prior to use.
- Neither the guidewire introducer or the torque device are intended to enter the body.
- To prevent damage to the guidewire coating, manipulate the guidewire using the torque device at proximal end of the guidewire.
The .014” wire with distal polymer jacket is not DMSO compatible.
The guidewire is compatible with other ancillary products used in intravascular procedures. Any over-the-wire style of micro catheter used with the guidewire should have at least 0.0025” (0.06 mm) clearance between the lumen of catheter and the guidewire.
Procedures requiring percutaneous catheter introduction should not be attempted by physicians unfamiliar with the possible complications, which may occur during or after the procedure.
Possible complications include, but are not limited to, the following:
- Perforation of vessel or arterial wall
- Thrombus formation
- Puncture site hematoma
This device is intended for single use only. Discard the product after use. Structural integrity and/or function may be impaired through reuse or cleaning.
Never advance or withdraw the guidewire against resistance until the cause of the resistance is determined by fluoroscopy.
Do not attempt to move the guidewire without observing the resultant tip response.
See the device manual for detailed information regarding the instructions for use, indications, contraindications, warnings, precautions, and potential adverse events.
Do you need support for procedures? Or information on our products and solutions?
Our team is happy to help answer any questions you may have.CONTACT US